ASTM D is a standard test method for vapor pressure of petroleum products. Find out more on ASTM D and their application and eralytics products. Lab Instruments · Application · Brand · Industry · Measurement · Method · ASTM · ASTM D36 · ASTM D56 · ASTM D86 · ASTM D92 · ASTM D93 · ASTM D Due to the increased use of ethanol in fuel the measurement of the Reid vapour pressure (RVP) is replaced by ASTM D in the EPA “Tier 3.
|Published (Last):||9 September 2004|
|PDF File Size:||9.53 Mb|
|ePub File Size:||8.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
Mercury, or its vapor, has been demonstrated to be hazardous to health and corrosive to materials. A pressure transducer is measuring the resulting rise d51191 pressure.
Atmospheric distillation unit offers perfection from Link to Active This link will always route to the current Active version of the standard.
If a haze is observed init shall be indicated in the reporting of results.
The precision and bias statements for hazy samples have not been determined see Note And What is it Used For? Is Fracking the Way Forward? Work Item s – proposed revisions of this standard. When performing reference measurements using pentane, this effect was not visible.
The Setavap 2 is used for the measurement of gasoline, solvents, light crude oils, and similar products, using the “Mini” method. Dec 21 Read Regulatory changes and increasing safety concerns in recent asgm led to extensive adjustments in process and laboratory vapour pressure testing. Characterisation of Hydrocarbons in Shale R Trending Articles Risks and side effects The underestimated dang During the measurement a single expansion is performed, thus just the total vapour pressure a combination of the partial pressure of the dissolved air and the vapour pressure of the liquid is accessible.
Setavap 2 Vapour Pressure Tester
The valves utilise solvent resistant Kalrez seals. This shows that a small loss of volatile content happened during sampling.
The process monitoring of the vapour pressure during blending is capable of performing trustfully measurements without the need of ongoing sampling for laboratory measurement. Due to transport and storage some of the light ends possibly were lost before the measurement, thus lowering the resulting vapour pressure, compared to literature values.
New FM approved liquid analyser system is safe for h Work Item s – proposed revisions of this standard.
All valve operations are automated including sample draining also chamber venting and purging. The pressure is automatically measured at one minute intervals, until three readings are within 0. For specific safety warning statements, see 7. Historical Version s – view previous versions of r5191.
The calculated DVPE very closely approximates the dry vapor pressure that would be obtained on the same material when tested by Test Method D Alternatively, the test chamber can have a moveable piston that expands the volume after the sample introduction — the internal volume of which is five times that of the total test specimen introduced into the chamber.
Calibration and axtm equipment available. Users must determine legality of sales in their location. The inch-pound units given in parentheses are provided for information only. These steps are necessary for reproducible measurements. In between the multiple saturation steps it is required to let the sample cool again for approximately 2 minutes. No account is made for dissolved water in the sample.
ASTM D – 13 Standard Test Method for Vapor Pressure of Petroleum Products (Mini Method)
Dec 22 Read Measurements are made on liquid sample sizes in the range from 1 to 10 mL.
Vapour pressure is one of the key parameters measured in the gasoline blending process. Also EN can be considered as an equivalent to this method. Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the standard. It incorporates a pressure sensor and is used to create the vacuum above the sample to determine the vapour pressure.
This test method is suitable for calculation of the DVPE of gasoline and gasoline-oxygenate blends by means of a correlation equation. When dealing with the vapour pressure of crude oils recent incidents showed that the currently used measurement methods are not sufficiently reflecting the conditions that occur during transportation or storage.
This eliminates the need of a vacuum pump before or during the measurement. A facility is included to allow two part tests Pabs to be conducted at The online and laboratory vapour pressure measurements of gasoline were performed at